3-17. TDS MONITOR MAINTENANCE.
This task consists of:
General Mechanics Tool Kit (Appendix B, Section III, Item 4)
O-ring (2) - 5262
ROWPU shut down (TM 10-4610-240-10).
General Safety Instructions
High voltages in this equipment can cause serious injury or death. Be certain that
all power is removed before performing maintenance.
This procedure is the same for both cables except as noted.
Disconnect cable assemblies (6) and (7) from receptacles (11) and (12).
Loosen two screws (9) and open cover (8).
Pull four snap locks (1) and remove meter cover (2).
For cable assembly (7), go to step d.
For cable assembly (6), go to step e.
Tag and disconnect three wires from TDS monitor (10).
Tag and disconnect five wires from TDS monitor (10).
Remove locknut (3) and O-ring (4) from connector (5).
Remove adapter (5) from cable assembly