Remove the evaporator cover from the still and inspect the heating elements from the inside of the evaporator.
Warped or split elements should be replaced.
To remove an Inoperative or damaged heater, loosen the screw under the heater and push out the heater. If the
heater is badly scaled, remove the retaining nut and rock the heater back and forth (from inside the evaporator) to break
Install a new heater in place of the old heater. Tighten the screw only enough to hold the heater in place. This will
allow you to rotate the heater when installing the bus bars.
Install the bus bars and wiring between heater terminals. Retighten the heater retaining screws to eliminate the
possibility of leakage.
Add water to the evaporator and ensure that none of the heating elements are leaking.
Reassemble the still and connect the still to the electrical service.