TEST AND REMEDY
DROP IN PURITY
Inspect evaporator for scale and clean as
Test distillate for C02, NH3, etc. Re-
TIES IN FEED
duce water flow rate to help eliminate
volatiles from condenser.
Inspect evaporator for excess scale and
clean as required.
TION OF DISTILLATE
-LEAK IN CONDENSER
Test condenser for leaks and replace if
Inspect the suspected parts. Clean with
GROWTH IN DIS-
formaldehyde if required. See "Biologi-
cal Cleaning" under CLEANING METHODS.
DROP IN DISTILLATE
-EXCESS SCALE ON
Inspect and clean as required.
Check voltage. If less than 5-percent of
the rated voltage, notify electrician.
-LEAK IN DRAIN
Ensure drain valve is closed and not
-DAMAGED OR IN-
Check wiring, connections, etc. Test
heaters and replace if necessary.
-LOW WATER FLOW
Readjust throttling valve.
LOW WATER PRES-
Check water pressure. It should be at
-EXCESS SCALE IN
Inspect condenser for scale.
Test condenser for leaks. Repair or re-
place as required.
Inspect distillate line for any restric-
tions or trapping.
GENERAL. This section contains parts list information for the still. When ordering parts, ensure that the proper part
number and quantity are specified. Also, provide the serial number of the still.