TM 09241 B-12&P
2-10. OPERATION OF TURBIDITY TEST SET.
Open case and carefully remove equipment and supplies as needed. Note location for repacking.
An indent is provided on probe to align and secure deflector cone thumb screw.
Set up the meter.
Remove protective cover (1) and attach deflector cone (2) onto probe (3) secure into indent with thumb screw
Connect probe (3) lead to turbidity meter (5).
Observe correct polarity when installing battery. Incorrect connection can damage the equipment.
Do not use AC to DC adapter if a battery is installed in meter. Damage to the battery or adapter can result.
Connect AC to DC adapter (6) or remove four screws (7) and back cover (8), install a battery (9), and attach
back cover onto turbidity meter (5) as appropriate.
Figure 2-8. Turbidity Test Set