Turbidity can be monitored using the portable turbidity meter. Obtain the media filter outlet sample (4, Figure 2-6).
A portable flowrate indicator (1, Figure 2-80) is provided as a replacement for the panel flowmeters. This portable
flowrate indicator is stored in the special tools and indicators storage box no. 4 (Figure 2-24). Simply connect the
flowrate indicator to the end of the product to storage hose (2, Figure 2-80). Holding the flowrate indicator in a
horizontal position with the indicator on top, read the flowrate directly in gallons per minute. Make necessary
adjustments in the ROWPU to attain the desired performance.
There must be full pipe flow at the end of the indicator. If not, elevate the end of the pipe to
remove air and then return to the horizontal position and read flowrate. If, after this
procedure, full flow is not maintained, submerge the flowrate indicator in the storage tank in
a horizontal position with the indicator extending above the water surface.
Figure 2-80. Use of Portable Flowrate Indicator
Emergency Hasty Decon Preparation for Movement.
Wear protective gear when performing emergency decon. Failure to do so may lead to
Do not use DS-2 dawn solution.
Hasty decon. In an emergency, limited hasty decon, can be accomplished by mixing a solution of 8 gallons product water,
one pound RO element cleaning detergent, and 6.5 pounds calcium hypochlorite (six and one half packets).
Change 12 2-167