TDS MONITOR MAINTENANCE - continued.
This procedure is the same for both cables except as noted.
Install adapter (5) and O-ring (4) on cable assembly (6 or 7).
Position cable assembly in place and install locknut (3). Refer to wiring schematic in Appendix F
for cable assembly being repaired.
For cable assembly (7), go to step c
For cable assembly (6), go to step d.
Connect three wires as tagged to TDS monitor (10).
Connect five wires as tagged to TDS monitor (10)
Position meter cover (2) in place and push in snap locks (1)
Close cover (8) and secure with two screws (9).
Connect cable assemblies (6) and (7) to receptacles (11) and (12).
Calibrate and test TDS monitor (para. 2-31).